ProductsMeter systems Lancing tools Data monitoring Apps & software regulating Diabetes SUPPORTMeter solution
The mmsanotherstage2019.com Performa blood glucose meter makes testing your blood sugar quick and easy. It’s ready to use ideal out the the box, requiring no set-up. Every meter offers mmsanotherstage2019.com Performa check strips and also an mmsanotherstage2019.com lancing device. Valuable alerts remind you to check before and also after meals, offering you valuable information to assistance therapy decisions because that a healthy energetic life.
You are watching: Accu chek e-9 error
After 2 minutes undisturbed v strip inserted, or if powered on but left undisturbed because that 2 minutes
18 month after manufacturing date; check strips stay stable up to expiry day printed ~ above test strip vial, also after opened (test piece container need to be closed tightly after each test piece is removed)
How to use the mmsanotherstage2019.com Performa
Get precise results in 3 steps:Insert your test strip right into the meter and it turns on automatically.Apply your blood sample to the easy-to-see yellow window. Just 1 little drop is needed.In about 5 seconds, your test results will show up in the easy-to-read display.
This website has information on products which are targeted come a wide variety of audiences and also could save on computer product details or information otherwise not easily accessible or precious in your country. Please be mindful that we do not take any responsibility because that accessing such information which may not comply with any kind of valid legitimate process, regulation, registration or consumption in the country of your origin.
Battery is dead. Insert brand-new battery. Insert one 3-volt lithium battery (coin cell type CR2032)Display is damaged. Call (insert affiliate contact information)Meter is defective. Contact (insert affiliate contact information)Extreme temperatures. Relocate the meter come a much more temperate area.
Your meter actions blood glucose within a defined selection (0.6–33.3 mmol/L).The screen may indicate that her blood glucose is over than the measuring variety of the device (33.3 mmol/L).Immediately repeat the test. If your test result matches just how you feel, call your healthcare professional immediately.If this go not enhance how you feel, contact.
Your meter actions blood glucose in ~ a defined variety (0.6–33.3 mmol/L).The display screen may suggest that her blood glucose is reduced than the measuring variety of the mechanism (0.6 mmol/L).Immediately repeat the test. If your test an outcome matches how you feel, contact your healthcare professional immediately.If this go not enhance how you feel, contact.
The test strip might be damaged or not effectively inserted. Remove and reinsert the check strip, or change it if damaged.
Your blood glucose might be extremely high or a meter or a test piece error has occurred.If your test result matches how you feel, contact your healthcare expert immediately.If her test result does not enhance how girlfriend feel, repeat the blood glucose test.If the E-3 code still appears for her blood glucose test, her blood glucose result may be very high and over the system’s analysis range. Call your healthcare skilled immediately.If the second test an outcome does not enhance how girlfriend feel, do a manage test v the control solution and a brand-new test strip.If the control an outcome is in ~ the acceptable range, review the suitable testing procedure and also repeat the blood glucose test v a new test strip.If the control an outcome is not within the agree range, contact Roche
Not enough blood or regulate solution was attracted into the test strip for measure up or was applied after the test had actually started. Discard the check strip and also repeat the blood glucose or regulate test.
Blood or regulate solution was used to the test strip before the flashing fall symbol appeared on the display. Discard the test strip and also repeat the blood glucose or control test.
An digital error occurred, or in rarely cases, a used test strip was removed and reinserted. Turn the meter o. And on, or take it the battery out for 20 seconds and reinsert it. Perform a blood glucose or manage test.
The temperature is above or below the proper selection for the system. Refer to the test piece package insert for system operating conditions. Move to one area with the ideal conditions, wait until the meter and also test piece reach the operation temperature (8-44 °C), and repeat the blood glucose or regulate test. Do not artificially heat or cool the meter.
The battery is virtually out of power. Change the battery now. If the message reappears after ~ the battery has actually been replaced, remove the battery again, press any type of meter button, climate reinsert the battery.
See more: What Writing Characteristic Determines That A Biography Is An Autobiography
The time and also date settings may be incorrect. Make certain the time and also date are correct and also adjust, if necessary.